How to Clean a Rudolph Research Density Meter: DDM2911
Trouble viewing this video? Click Here.
Technical video for the Rudolph DDM 2911 Density Meter. This video details the procedure required to clean the Density Meter prior to or after making a liquid density measurement.
The video covers using a syringe to inject cleaning solvent, drying agent, capturing waste, and drying the instrument’s U-Tube with the on-board air pump. Using VideoView TM and simple test measurement the user can verify that the instrument is clean, dry, and ready for a new liquid density measurement.